The first KREBS viscometer totally automatic. No need to lower any handle!
Instrument based on the traditional KREBS method: a constant unique speed (200 rpm) and a unique rotating spindle. Microprocessor converts automatically measurement that is displayed continuously in the select unit: KU, g o cP. The VK 2000 allows user to choose between 2 different operational methods, manual and automatic. If the Manual Mode is selected, user can choose any kind of container to perform measurements.
In Manual Mode, UP and DOWN keys move spindle to required position. Press START and STOP to begin with measurement and to stop rotation of motor. Automatic Mode allows to select and modify through keyboard 2 parameters: SampleWaitingTime -SWT (time in which spindle remains immersed in sample before rotation begins) and SampleMeasuringTime –SMT- (rotation time of the spindle immersed in sample).
MYR KREBS viscometer VK2000 is used to measure viscosity in paints, coatings, inks, adhesives and pastes both in production process and laboratory quality control.
|KU||40,2- 141||0,1 KU
|G||32- 1.099||1,0 G
|CP||27 -5.274||5 cP
|Accuracy||± 1% of full scale (FSR)|
|Operating temperature||+10ºC - +40º|
|Protection classification||IP 20|
|Viscosity reading||KU, g, o cP|
|Spindle||standard KREBS-type spindle
|Unidirectional interface RS232||Used together with a serial printer delivers a printed ticket meeting requirements of the storage data in quality control
Basic delivery includes